Preparing HDSTM Personal Monitors for field sampling just became a lot easier with the HDSTM Personal Monitor Prep Station!
Simply attach an HDSTM Personal Monitor to the HDS-PS1 Prep Station and use a vacuum pump and cylinder containing Helium and BFT to evacuate, fill, evacuate, and fill the sampler. The personal monitor is now ready to be sent to the field. To clean samplers, simply evacuate, fill, and evacuate the samplers then place in an oven overnight at 70–100°C. Cool the samplers then follow the normal fill, evacuation, and fill process. Your samplers are now ready for the next field sampling event.
The HDS-PS1 comes with the prep station manifold as shown, and includes the compound gauge, 3-way valve, and female Micro-QTTM Valve. Diaphragm pump and cylinders containing the Helium/BFT mix are sold separately.